D CV. Read this owner’s manual thoroughly before use Owner’s operation manual for series. DT Series. 3 1/2 Digital Multimeter. Read this owner’s manual thoroughly before use. DT series multimeter have been designed according to IEC concerning electronic measuring. View and Download Histest DT owner’s operation manual online. DT Multimeter pdf manual download.
|Published (Last):||26 April 2012|
|PDF File Size:||3.70 Mb|
|ePub File Size:||4.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
Before attempting to open the case, always be sure that test leads have been disconnected from measurement circuits.
If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed at a nominal cost. Table of Contents Add to my manuals Add. To print the manual completely, please, download it. This function allows the meter to output a signal as a signal generator.
Enter text from picture: Use the proper terminals, function, and range for your measurements. Got it, continue to print. Related Manuals for histest DT No related manuals.
Front Panel Description This function allows the meter to output a signal as a signal generator. Don’t show me this message again. A coupling capacitor should be used when connecting the meter to a circuit. Safety Information If the defect has been caused by a misuse or abnormal operation conditions, the repair will manule billed at a nominal cost.
Before measuring current, check the meter’s fuses and turn power off to the circuit before connecting the meter to the circuit. Set the range switch to the desired range.
DTD Manual Ranging Digital Multimeter
To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than 15 minutes. Page of 13 Go. Operating Instruction “COM” jack.
If the resistor to be measured is connected to mabuale circuit, disconnect circuit power and discharge all capacitors before measuring resistance. Don’t have an account?
DT-830D Manual Ranging Digital Multimeter
Close case and tighten screws completely before using the meter to avoid electrical shock hazard. Disconnect circuit power and discharge all high-voltage capacitors before testing resistance, continuity, diodes, or capacitance.
Page 6 To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than 15 minutes.
A test signal will appear between the and “COM” jacks. Set the range switch to ” ” or ” ” range. Battery And Fuse Replacement Before attempting to open the case, always be sure that test leads have been disconnected from measurement circuits.
By pressing ‘print’ button you will print only current page. The voltage is approximate 5Vp-p. dt830x
In the latter case, the transient stresses are variable; for that reason, its necessary that the transient-withstand -capability of equipment is made known to the user. Comments to this Manuals Your Name.